Payment & Shipping Terms:
|Power Supply:||110VAC±10％/220VAC±10％||Available For TMPs:||TMP Pumping Speed 600-2000 L/s|
vacuum system controller,
vacuum pressure controller
Power supply & Controller , TCDP-II , Multi-function & Low noise , LED
TCDP-II series controller of molecular pump has a top performance on multi-function and low noise. It can be applied to various molecular pumps. This user manual provides with precautions regarding installation, commissioning, operation, trouble diagnosis and daily maintenance. Please read it carefully before any installation and usage is commenced to ensure a proper operation. This manual shall be available together with the controller, and should be well kept for future reference and convenience of maintenance.
|Input Frequency(Hz)||50/60 ±5％|
|Max output power||750|
|Output voltage( V/AC)||no more than 66|
|Operation current (A)||no more than 4|
|Max Starting current(A)||no more than 14|
|Start/Stop||Available with Key panel , external control terminals and computer|
|Display||In the front panel: Output frequency, Output current, Output
voltage, error type and function code, operation code and TMP model,
2 separate LED lights indicate the states of Electrification and Error.
|Protection||In case of undervoltage, overvoltage, overcurrent, overheat, overload,
current output limit, overcurrent and overheat protection,
Password error, external interference protection
|Working environment||Facility environment: no highly corrosive gases and dust,
Humidity:＜90% ( no condensation),
Vibration Strength: ＜0.5g (gravity acceleration)
1. Away from rain, drips, steam, dust and oil-dust; no corrosion, flammable gas, liquid; no metal particles or metal powder, etc.
2. Ambient temperature: within range of -10℃~+50℃.
3. Relative humidity of environment shall be lower than 90%, without any bead coagulation.
4. No strong electromagnetic interference.
5. Vibration strength: below 0.5g (acceleration).
6. Should The controller should be installed inside control cabinet, sufficient ventilation shall be ensured. Ventilation space shall be secured in the surrounding
Contact Person: sales